IEC 60529 IPX1 IPX2 Movable Vertical Rain Drip Box Waterproof Test
Vertical Drip Rain Testing Machine is designed and manufactured
according to IEC 60529 Degrees of protection provided by enclosures
(IP Code), this equipment is widely used in lamps, electric and
electronic products certification testing of IPX1,IPX2. This device
has been widely adopted by domestic and foreign products
certification laboratories, the performance of this equipment
reaches to the international advanced level.
The program of this test equipment is humanized, users can process
IPX1~2 testing without re-setting any parameters manually by
choosing the default test options; user also can set the test data
to meet different testing standards by choosing the refine test
The water-flow rates is ＞10L/min ±5%, pure water and without
impurities, equipped with water cleaning filter
Waterproof test range
Vertical drip rain area
Drip needle aperture
Distance between holes
Dripping speed (water flow)
For the purpose of the tests, the surface area of the enclosure is
calculated with a tolerance of 10%.
Adequate safety precautions should be taken when testing the
equipment in the energized condition.
Test for second characteristic numeral 1 with the drip box
The test is made with a device which produces a uniform flow of
water drops over the whole area of the enclosure.
The turntable on which the enclosure is placed has a rotation speed
of 1 r/min and the eccentricity (distance between turntable axis
and specimen axis) is approximately 100 mm.
The enclosure under test is placed in its normal operating position
under the drip box, the base of which is larger than that of the
enclosure. Except for enclosures designed for wall or ceiling
mounting, the support for the enclosure under test should be
smaller than the base of the enclosure.
An enclosure normally fixed to a wall or ceiling is fixed in its
normal position of use to a wooden board having dimensions which
are equal to those of that surface of the enclosure which is in
contact with the wall or ceiling when the enclosure is mounted as
The duration of test is 10 min.
Note: When the base of the drip box is smaller than that of the
enclosure under test, the latter may be divided into several
sections, the area of each section being large enough to be covered
by the dripping water. The test is continued until the whole area
of the enclosure has sprinkled for the specified time.
Function interface of this tester: